FIELD SERVICE BULLETIN NO. 25
SBE 56 Temperature Loggers
Print version Link to all Field Service Bulletins
This field service bulletin applies to SBE 56s manufactured and delivered before 2012 ‑ Serial Numbers (SN) < 1088.
The SBE 56 uses a Saft LS14500 battery. A few customers have reported that battery endurance was significantly below the value estimated by SeatermUSB software. Investigation and thorough testing at Sea-Bird, and extensive discussion with the battery manufacturer, has confirmed that the battery endurance could be reduced by up to 40% if all of the following apply:
Comparative testing at Sea-Bird of other bobbin-style AA lithium batteries from various manufacturers did not eliminate this problem; all of these batteries had similar characteristics.
NOTE: For most typical customer applications, the SBE 56 battery endurance is sufficient, even with the potential 40% reduction. For example, the calculated endurance for a 15-second sample interval is 717 days (≈ 2 years). With a potential 40% reduction in battery capacity, the SBE 56 could be deployed for 430 days (≈ 1.2 years).
If any one of the described conditions is removed, the battery will regain its expected endurance. For example, if you change the orientation, the SBE 56 will begin sampling again and should provide the originally estimated number of samples.
The desiccant pills originally used in the SBE 56 were discontinued by all desiccant suppliers. The industry has moved to a larger capsule, which does not fit in the existing SBE 56 housing.
Affected SBE 56s:
If you require the full battery capacity calculated by SeatermUSB, and must deploy/redeploy the SBE 56 before returning it to Sea-Bird, and deploy it in a horizontal orientation, or thermistor end up (battery button end (+) up).
If requested on the Service Request Form when the
SBE 56 is returned for service/
calibration, Sea-Bird will change the battery cradle orientation in the housing, so that for the typical deployment (thermistor end down) the battery button end (+) is up.
SBE 56s manufactured in 2012 and later (SN > 1088):
Sea-Bird changed the battery cradle orientation in the housing, so that for the typical deployment (thermistor end down) the battery button end (+) is up.
Sea-Bird extended the housing length by approximately 1 cm to accommodate the larger desiccant capsule. When the SBE 56 is returned for service/calibration, we will replace the housing with the longer housing.
Sea-Bird Home Phone: (+1) 425-643-9866 E-mail: email@example.com